Cursor 1 with measuring line
Test Instrumen s DSO
Auto level mode
Shifting of cursor 1 (fine adjustment)
Cursor 2 with measuring line
Test Instruments DSO
Shifting of cursor 2 (fine adjustment)
Amplitude channel A
Amplitude difference A
By pressing the button "Freeze frame", the cyclic repetition of the measurement can be stopped or continued.
The buttons "Cursor 1" and "Cursor 2" are dis- played in the freeze frame function. The voltage values of the curves can then be displayed sepa- rately for each channel.
The associated voltage value of each curve can be determined by shifting the cursor position.
The time value represents the distance from the trigger point "T".
The difference in amplitude is displayed auto- matically if both "Cursor" buttons are selected.
The difference in amplitude results from the dif- ference between the voltage value of the curve at the position of cursor 1 and the voltage value at the position of cursor 2.
The difference in time is the interval from cursor 1 to cursor 2.
If both cursors are selected, only cursor 2 can be shifted on the curve.